Message 8 of If it fails even using the 2m cable, troubleshoot the test station. Do you have more than one GPIB controller? Hello, I’ve found only the A driver. As you have probably already determined, the A and D versions have the same programming manual and aside from the differences of the output range, they would appear to be compatible.
|Date Added:||11 December 2010|
|File Size:||27.28 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I just wanted to make sure since Agilent is so certain it is not the instrument. I took a quick look at the user manual for the series and it appears that you should have no problem using this driver with the D.
GPIB-USB-HS and C++builder sample code problem – Discussion Forums – National Instruments
Also, I tried other instruments on the same cables and the communication was fine. Oabview think this is highly unlikely. However if the issue doesn’t go away, it probably is something that agilent would be able to answer better. Infiniium scopes and InfiniiMax differential probe amplifiers have a unique method of applying DC offset which considers whether the signal being probed is single-ended or differential That is a terrible answer since I am not a big fan of “pointing fingers”, but to me that seems like the only logical conclusion.
What is the measurement setup for obtaining same spectral levels of measured signal or noise irrespective of the sweep frequency span of the A?
Luca has a great suggestion. What should I do to troubleshoot the instrument? Have you check on Agilents website?
I am getting the error when I manually send the command using MAX. Message 9 of Please help me ,if anyone has this driver please email it to me raj klsi. Message 5 of These types of measurements can be made using long average times….
Message 4 of 5. I wonder if there is a difference from the A version to the D version. Slotless contacts make electrical connection with the mating male pin without changing the outer diameter of lsbview female center conductor. You might want to issue the SYST: The current version of firmware can make gated measurements with a total time of ms.
Message 8 of If so, then it is probably an instrument issue where the connection could be broken or something to that the data you send is not what it receives.
I am using EB by Agilent and 864d am getting the error: When I send the reset command, it says invalid character. Something so that the string you 8648x sending to the device is not the one that is received. Thank you again for all your help.
Agilent 8648 Signal Generator
It labvew definately connected correctly. How do I execute? Precision slotless female contacts are used to reduce the reflection uncertainty at a connector interface.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. For the InfiniiMax probes, what effect does the “Signal being probed” setting have? I got the same error. Any update on this one tried the wire ID quiery to false, didn’t work. Message 1 of 5.